The Mask Lab disposable mask passed our particulate filtration efficiency machine test with a score of 93.04%. This 3-ply blue surgical mask has ear loops instead of head straps or ties.
Mask breathability is determined by airflow resistance, which we measure in pascals. The target for adults should be around 150, and the target for children should be close to 100. The airflow resistance of this mask is 78.80.
This mask has a cardboard odor. We can't recommend it as it's missed the filtration claims on its packaging and features shoddy construction.We found this mask on Amazon. Its country of origin is India. Check out our video review for more details.